Following successful cuff deflation, the patient’s clinical condition and the type of tracheostomy tube in situ will dictate the next stage in the process of weaning. The following sections describe different weaning methods.
NB The cuff deflation guideline should be carried out prior to completion of ALL methods of weaning as described in the following sections.
A number of different methods of weaning exist. Consideration should be given to individual patients to ensure that the safest and most appropriate method of weaning is identified.
Weaning times can vary due to the original reason for insertion of the tracheostomy tube and length of time on mechanical ventilation. Once the following has been achieved weaning can commence.
Weaning from a tracheostomy tube by either decannulation cap or speaking valve increases the workload of breathing. Gradually increasing this workload will help build respiratory muscle to ensure that following decannulation the patient is able to breathe and clear secretions effectively. Weaning can commence after the patient has passed a cuff deflation trial. The process of weaning must be planned and clearly documented (Weaning record of care/Appendix 2).
Weaning should not occur overnight, speaking valves or decannulation caps should be removed as complications can be more difficult to identify.
Double lumen uncuffed trachstomy tube with fnestrated inner lumen tube and speaking valve (shown apart)
Double lumen uncuffed trachstomy showing fenestrated inner luman and speaking valve insitu
During the weaning process, one of the benefits that can be achieved for the patient is the ability to vocalise for short periods of time. Adjuncts which allow patients to vocalise increase the workload of breathing and therefore should only be considered for use in patients in whom it has been agreed within the multidisciplinary team to commence weaning from the tracheostomy tube.
Please note using a decannulation devices involves partially or completely occulsion of the tracheostomy tube by the decannulation device, if there is any degree of airway occlusion from oedema, of the cuff is not completely deflated or a non-fenstrated tracheostomy tube is insitu, the patient may suffer complete airway obstruction. Prior to attatching the device, an assessment of airflow via oro/nasophaynx is recommended, this can be assess by lightly applying finger occlusion to the tracheostomy tube prior to attatching the device to observe for any signs of airway obstruction.
Weaning with a Single Lumen Tracheostomy Tube
Suitable for paients who are able to wean over a short period of time
Click here for instruction on weaning patients using a single lumen tracheostomy tube
Weaning from a Decannulation Cap
Suitable for patients with a fenstrated double lumen tracheostomy tube. The cuff MUST be deflated.
Click here for instruction on weaning patients using a double luemn tracheostomy tube and a decannulation cap
Weaning with a Speaking Valve
Suitable for patients with a fenstrated double lumen tracheostomy tube. The cuff MUST be deflated
Click here for instruction on weaning patients using a double luemn tracheostomy tube and a speaking valve
Reasons for continuation of tracheostomy
Click here for instruction on the process of decannulation